E-light applies three core technology, IPL, RF, and epidermis cooling. It combines the advantage of IPL and RF, in one hand, the skin absorbs IPL energy selectively, and the skin also absorbs RF energy, on the other hand, the RF-surface-touch cooling technology removing the discomfort and complications by high heat of IPL. Therefore, the effect and safety of E-light makes a breakthrough. 1. Principle Applying the technique of IPL and Bi-polar Radio Frequency, the equipment acts on the entire skin corium and connective tissue, stimulates the collagens of different depth to realign and grow, then gets treatment effect. Bio-polar RF release energy at the same time, with the selective absorption for the skin to light energy, various pathological tissues of corium and epidermis creates Photothermolysis effect. Since light energy is far below conventional IPL, the pigment tissue and vascular pathological tissue of the corium and epidermis would absorb more energy without injuring the epidermis, and then remove these pathological tissues better and get the effect. 2. Application： 1. Hair removal (including dark skin); 2. Skin rejuvenation 3. Acne treatment; 4. Removing freckle, age spots, sun spots, pigmentation troubles, etc; 5. Removing red face, red capillary, and bottle nose
IPL Power: 800W ● RF Intense Electric Energy: 5MHz ● Light Energy: 15-50J/cm2 ● Pulse Amount: 1-6 ● Pulse Width: 1-10 ms ● Pulse Delay: 1-100 ms ● Repeat frequency: 1-4s ● RF working time: 50-2500 ms ● Spot size: 8x40mm ● N.W.: 50kg ● Voltage: 230V±10%, 50~60Hz
Connect the handle and filters: 1. Put the E-light handle on the hanger 2. Connect the E-light handle to the connecting port on the back board of the machine tightly. 3 Check the water holes of the backboard, make sure the water outlet hole is plugged well.
Water infusing: 1 pull out the plug of the water inlet hole，insert the water funnel 2 infuse water till water spill out from the spillway hole.
Checking 1 Make sure there is no water leaking; .2 Make sure there is no moisture or wet for electricity connecting parts; 3 Make sure the key switch is in the off position; 4 Make sure the emergency stop switch is in the on position.
Starting the equipment 1 Connect the power cord, turn on the key switch; .2 Double confirm there is no leaking of water circulation, fold the protective cover by turning it right.
Debugging 1. Make sure water and electric circuit is correct connected; then turn on the power to start the machine. 2. Turn on the master power switch on the backboard; 3.Turn on the emergency switch (clockwise turn) in the front side of the mainframe
Your review is awaiting approval
There are no reviews yet.
Be the first to review “3 In 1 Ultrasonic Cavitation 40K Weight Loss
Your email address will not be published. Required fields are