Adopting the Q-Switch ND YAG model,the equipment uses YAG to emit high energy instantaneously which smashing the pigment of pathological tissues,namely the YAG leading explosive effect:High energy emits instantaneously,making particular wavelength penetrating epidermis and getting into pathological pigment tissues in only 6 nms. Then pigments expand in top speed and explode.The superficial epidermis would be bounded our of body;other part of the pigments would split into granules which would be engulfed by macrophages,then exclude by the lymphatic system finally.The pigment becomes shallow gradually finally disappear while the surrounding normal skin tissues keep sound as they do not absorb particular wavelength .
1.Pigment skin pathological changes,pigmentation caused by color pigment mixture,corium spot,freckle,mole,birthmark,nevus of Ota,etc.
2.Specialized in removing red capillary,coffee,brown and other colorized tattoos.
3.Effectively remove all kinds of embroider eyebrow,soak lip,eye line,and lip line.
The system gem-adjusting Q switch can effectively cure the mixed pigments by pigment precipitation of extraneous source with pathological pigment change of endogenous source, inclusive of erroneous dark eyebrow-embroidery, lip-embroidery, lip-soak, eye line-embroidery, lip-line, tattoo (inclusive of tattoo by extraneous wounds), pigment precipitation by sensitive source, freckle on female’s face (male’s), coffee spot, old age spot, black mole, and birthmark in black, blue, purple, brown, red and etc. Operation course for different illness case as follows:
1. Operation for tattoo in eyebrow
Customer could not open eyes during Operation, operator should adjust energy according to depth of tattoo color, hold the handle and move it to Operation area slowly to make customer feel acceptable in mentally. Right now, the speed of flash could be set 3HZ, keep head of the handle a two-finger distance to Operation area, start Operation from head of eyebrow to end of eyebrow. It is needed to treat again in 5-10 minutes if the color of tattoo re-appear.
2. Remove eye line
The speed of flashes could be set 4-5HZ,adjust the energy of light to minimum energy, keep a one-finger distance between head of handle and Operation area. Operator should strictly protect customers eyes with panel for eyelid or medical pledget (panel for eyelid)of 1cm depth.
Operation should be started from down eye line, customer should deeply close eyes and keep eyeball up-sight inside. Operator should press the pledget (panel for eyelid) with forefinger and push up eyelid, peel down eyelid with middle finger to bare the eye line, treat it with flashes of 4-5Hz. Then, treat up eye line. Also customer should close eyes and keep eyeball down-sight inside. Operator should press the pledget (panel for eyelid) with middle finger and push eyelid down, push eyelid up with forefinger to bare the eye line, treat it with flashes of 4-5Hz.
3. Remove lip line
The speed of flashes could be set 5Hz,keep a 3/4 finger distance between head of handle and Operation area. It is better to remove black, coffee, deep, red and brown lip line with 1064 head at first time and then, treat again with 532 head.
Notice: All tattoos in lip should be treated in 6 months after making. OR removal Operation will fail. It should have 45-60 days interval to continue next Operation until remove all.
4. Remove tattoo
The speed of flashes could be set 3Hz/5Hz.It is better to use high energy for Operation until Operation area seeps a little blood. It should have 45-90 days interval to continue next Operation until remove all.
5. Remove spot and nevus
The speed of flashes could be set 3Hz/5Hz. Keep 3-4 fingers distance between head of handle and Operation area. It is better to use high energy for Operation until Operation area seeps a little blood. It is advised to use 1064 head for normal spot and nevus Operation, but for deep color spot and nevus, it is better to use 532 head for Operation. It should have a 15-30 days interval to continue next Operation until remove all.
6. Remove birthmark
The speed of flashes could be set 3Hz/5Hz.It is better to use high energy for Operation until Operation area seeps a little blood. It is advised to use 1064 head for birthmark Operation, but for red and deep color birthmark, it is better to use 532 head for Operation. It should have 45-90 days interval to continue
Cooling condition: wind system+closed circulating+cooling system
type: Q-switch Nd YAG With RED target light to aim at the skin.
The pulse energy: 0~2000mJ
Frequency:1-5Hz continuous adjustable
Tips: 1064 nm & 532 nm and black doll tips
1 x Main Unit
1 x Hand-piece
2 x Tips
1 x Black doll tip
1 x Water funnel
1 x Foot touch
1 x Protective goggle
2 x Goggles (1 for operator, 1 for customer)
1 x Power line (we stock 110V-240V power version and US.UK.UA.EU plugs for the country all of the world)